Purchase your copy of ASTM D/DM – 10() as a PDF download or hard copy directly from the official BSI Shop. All BSI British. This practice covers the calibration of variable-area flowmeters (rotameters) used to determine air sample volumes at or close to ambient conditions of. Find the most up-to-date version of ASTM D at Engineering
|Published (Last):||20 September 2006|
|PDF File Size:||15.14 Mb|
|ePub File Size:||12.14 Mb|
|Price:||Free* [*Free Regsitration Required]|
Since viscosity is propor- added by the wet test meter s3195 gasometer, assuming saturation, tional to the square root of absolute temperature independent by operating with the factor: Combining values from the two systems may result in non-conformance with the standard.
Subscription pricing is determined by: Last previous edition D — 73 1. Note the manometer reading when room air is used for calibrating gas.
ASTM D3195/D3195M – 10(2015)
Your Alert Profile lists the documents that will be monitored. A superscript epsilon indicates an editorial change since the last revision or reapproval. Already Subscribed to this document. Procedure Using Wet Test Meter 1 7. As the voice of the U. Add to Alert PDF.
This standard is not included awtm any packages. Repeat, going from high to low.
The values stated in each system may not be exact equivalents; therefore, each system shall be used independently of the other. Your comments are invited either for revision of this standard or for additional standards and should be addressed to ASTM International Headquarters. Proceed to Checkout Continue Shopping. This can be done logically in several ast. We have no amendments or corrections for this standard.
You may delete a document from your Alert Profile at any time. Individual reprints single or multiple copies of this standard may be obtained by contacting ASTM at the above address or at phonefaxor service e-mail ; or through the ASTM website.
ASTM D/DM – 10() – Standard Practice for Rotameter Calibration
This conforms to diaphragm type pump of adequate capacity and a ballast d395 of the ASTM methods for atmospheric sampling and volume or restrictor to eliminate pulsations. If you feel that your comments have not received a fair hearing you should make your views known to the ASTM Committee on Standards, at the address shown below.
Terminology See Test Method E You can download and open this file to your own computer but DRM prevents opening this file on another computer, including a networked server. Need more than one copy? Your comments will receive careful consideration at a meeting of the responsible technical committee, which you may attend.
Standard Practice for Rotameter Calibration
Please first log in with a verified email before subscribing to alerts. See bility of regulatory limitations prior to use. Procedure Using Gasometer room temperature, barometric pressure in accordance with Test 8.
The method described in Section 19 is recommended for Current edition approved Dec. Summary of Practice at or close to ambient conditions of pressure and temperature, in the aztm of atmospheres for pollutant content.
This standard is also available to be included in Standards Subscriptions. It is the 4. A number in parentheses indicates the year of last reapproval. Standards Subsctiption may be the perfect solution.
Higher accuracy is possible with the gasom- eous Fuel Samples2 eter.
ASTM D / DM – 10() Standard Practice for Rotameter Calibration
Use average values and meter water temperature for each meter reading. Use Methods D ast, Please first verify your email before subscribing to alerts.
Users of this standard are expressly advised that determination of the validity of any such patent rights, and the risk of infringement of such rights, are entirely their own responsibility. If the document is revised or amended, you will be notified by email. Scope ture scales are to be used when substituting values into the 1. Therefore, in order to prepare atm meter; room temperature for gasometer.